DIGITILT DATAMATE II DRIVER DETAILS:
|File Size:||13.3 MB|
|Supported systems:||Windows All|
|Price:||Free* (*Free Registration Required)|
DIGITILT DATAMATE II DRIVER
Please register, and get one. If you encounter problems with the direct download for these operating systems, please consult the driver download manager for the specific Digitilt Datamate Ii Usb model. When digitilt datamate ii readings are stable and ready to record, the DataMate displays a graphic ready signal. The operator records the reading by clicking a hand switch. Metric control cables are graduated with yellow marks at 0. There are numeric marks at 5-meter intervals. English control cables are graduated with yellow markers at 2-foot digitilt datamate ii and red marks at 5-foot intervals.
|fake taylormade r7||Related titles|
|konica c454||Ask a Question|
|how do i remove wildtangent web||Digitilt DataMate II Driver for Dell - Vostro 1500 working on Microsoft Windows XP Professional|
|netmos 9835 pci multi i o controller||DriverIdentifier Tool|
|a4tech pk-333mb webcam||Related titles|
There are numeric marks at foot intervals. In addition, there are yellow bands of tape at 10 foot intervals. Each band represents 10 feet from the last numeric mark. For example, 4 bands represent 40 feet from the last numeric depth mark. Accurate inclinometer measurements depend on consistent placement of the inclinometer probe. Always align the depth marks on the control cable with the same reference. We recommend using a pulley assembly to digitilt datamate ii with depth control. The jam cleat on the pulley assembly holds the cable and the top edge of the chassis provides a convenient reference for digitilt datamate ii depth marks.
- Hardware and Software Documentation
- Digitilt Datamate Ii Usb Driver Download
- Wizard Textware -- Hardware and software documentation
- Drivers >>> Digitilt DataMate II driver
- Why Use DriverIdentifier?
- Training Materials on: Inclinometer
The small pulley assembly is used with 48 mm and 70 mm casing 1. The large pulley assembly is used with 70 digitilt datamate ii and 85 mm casing 2.
Digitilt Datamate Ii Usb Driver Download
Remove the pulley from the chassis. Clamp the chassis to the top of the casing. Insert the inclinometer probe and control cable. Replace the pulley. Your data reduction software can automatically adjust for this, digitilt datamate ii keep your survey procedure simple: use the marks on the cable and the top digitilt datamate ii of the pulley chassis for reference. Let the software do any extra work required. Check that operators consistently use the pulley assembly.
If the pulley is digitilt datamate ii for one survey and not for the next, the resulting data sets will not be directly comparable. Sometimes a monument case or a protective pipe makes it impossible to attach the pulley assembly to the casing. In this case, you can make a removable adapter for the pulley assembly. If you use an adapter, be sure to use it consistently. Calibrate your Cable: If you have time, calibrate your cable, recording the exact.
DigiPro2 User Manual
Do not use solvents to clean the cable. Be sure the protective cap is in place before immersing the end of the cable in water.
Do not immerse the Lemo connector. Cleaning Connectors: If it digitilt datamate ii necessary to clean the connector, use a cotton swab. Do not use spray lubricants or electric contact cleaners. Solvents contained in such products will attack the neoprene inserts in the connectors. Drying Connectors: When you return to the office, remove protective caps from the. Allow connectors to air-dry well for a number of hours. Storage: Store cable on a cable reel when possible. The reel should have a mini.
If a reel is not available, use the technique below to coil the cable. Coiling Cable 1. Use the same probe and control cable for each survey, if possible.
Use a pulley assembly, if possible. It protects the control cable and provides a good reference. Use a consistent top reference.
If one technician uses a pulley and another technician does not, probe positioning will be inconsistent, and data will have to be manipulated before it is useful. Always draw the probe upward to the reading depth. If you accidentally draw the probe above the intended depth, lower the probe down to the previous depth, then draw it back up to the intended depth. What digitilt datamate ii the DataMate II?
The Digitilt DataMate II is a readout used to record surveys taken with Digitilt inclinometer probes and spiral sensors. Multi-Part Setup. This multipart setup should not be necessary for most systems.
Part 1: Installing the Digitilt DataMate USB Driver. Turn on your computer.